Pricing information and quote request on Geotechgate.com
The constant head permeability test involves flow of water through a column of cylindrical soil sample under the constant pressure difference. The test is carried out in the permeability cell, or permeameter, which can vary in size depending on the grain size of the tested material. The soil sample has a cylindrical form with its diameter being large enough in order to be representative of the tested soil. As a rule of thumb, the ratio of the cell diameter to the largest grain size diameter should be higher than 12 (Head 1982). The usual size of the cell often used for testing common sands is 75 mm diamater and 260 mm height between perforated plates. The testing apparatus is equipped with a adjustable constant head reservoir and an outlet reservoir which allows maintaining a constant head during the test. Water used for testing is de-aired water at constant temperature. The permeability cell is also equipped with a loading piston that can be used to apply constant axial stress to the sample during the test. Before starting the flow measurements, however, the soil sample is saturated. During the test, the amount of water flowing through the soil column is measured for given time intervals.
Knowing the height of the soil sample column L, the sample cross section A, and the constant pressure difference Δh, the volume of passing water Q, and the time interval ΔT, one can calculate the permeability of the sample as
K=QL / (A.Δh.Δt)
Can you edit this page? Send us an email to firstname.lastname@example.org
* The images and videos on this page are niegher owned nor hosted by Geotechdata.info. You are responsible for checking the copyright status of images with the owener before using them.